Monitoring Hydroelectric Dam Sumps
TD-4100 Oil in Water Monitors for Hydroelectric Dam Sumps
TD-4100 Oil in Water Monitors for Hydroelectric Dam Sumps Hydroelectric Dams produce electric power via the force of the water flowing through the dam. There are numerous sources for oil spillage: The hydroelectric turbines are lubricated by standard lubrication oils; The very large water control valves are normally hydraulic powered and are lubricated with various greases and oils; Vehicles including fork lifts, trucks, etc., can spill oils and fuels. Any accidental spillage of oil or grease or intentional wash down water containing minor amounts of oils, greases and detergents, must be collected and treated before discharge to the river. These fluids along with any water leaks are collected in sumps in the dam and are regularly pumped out. In many cases, the volume of water is substantial. To protect the river environment the sumps at some advanced locations have oil in water separation systems and oil in water monitors. The installation of the TD-4100 in these applications is very simple. All that is needed is a sample of the water at 2 gpm @ 5 psig and a way to drain the water back to the sump. A dry air supply is also needed to prevent fogging within the non-contact falling stream flow cell. See drawings and P&ID below. The TD-4100 with its UV Fluorescence technology and non-contact flow cell has very strong advantages over any other oil in water instrument for this application. They are:
- Very high sensitivity to oils in river water - 50 ppb is typical and as low as 10 ppb in some cases. No other oil in water monitor can claim this level of sensitivity. High sensitivity allows the end user to alarm well below discharge limits and make corrections to their pumping sequence or to their treatment system. Even other UV Fluorescence competition cannot detect below 1 ppm and light scatter instruments are worthless if there are no oil droplets to scatter the light.
- Very low sensitivity to suspended solids - Typical river water coupled with wash down and drainage water from the dam will have a fair amount of suspended solids. Suspended solids are the downfall of any light scatter instrument because they register a positive signal as oil even when there is no oil present. Note that the TD- 4100 can see as high as 800 NTU (turbidity units) with less than a 10% change in the reading from distilled water.
- Very low maintenance - the only maintenance issue is very fine silt that may settle in the pipe-work. This includes the pipe-work feeding the instrument as well as the bubbletrap. Our customers have found that a periodic flush of the lines with filtered fresh water is sufficient to keep the lines maintenance free.
- Sample pour through feature - Using the standard supplied calibration bottles, the end user can take water samples from various locations within the dam structure to provide instant check of the presence of oil. The water samples are simply poured through the bubbletrap and the concentration is displayed on the screen.
The pictures below are from a recent visit at the US Army Corps of Engineers dams on the Columbia River in Oregon. At this time they have one TD-4100 installed on "The Dalles" dam and one on order for the Bonneville Dam. We also have instruments installed with BC Hydro in Vancouver, City of Mt. Holyoke, MA Hydro Plant, Danube Hydro in Austria and many others worldwide.
The Dalles Dam - Oregon, USA - USACE
TD-4100 installed after Mycelx oil removal filters
The Dalles Dam Oregon, USA - USACE
TD-4100 Oil in Water Monitor for Discharge to the Columbia River The Dalles Dam Oregon, USA - USACE